Hospitals are facing a flood of respiratory failure patients due to the COVID19 virus. Respiratory therapies such as nasal high flow (NHF) oxygen may provide clinical benefits to some patients with hypoxemic respiratory failure from COVID19 infection.
However, clinicians are reluctant to use NHF in these patients due to the perceived risk of viral transmission to healthcare workers via virus-laden droplets or aerosols.
We introduce the vacuum exhausted isolation locker (VEIL), in clinical use as a 180 L enclosure to extract exhaled aerosols and droplets from the patient and provide a safety barrier for hospital staff. Exhaust tubing connected to hospital suction via viral particle filters continually extracts air at 60-80 L/min, sustainable from patient room suction taps as U.S. hospital construction code (NFPA99) requires >85 L/min. With a patient’s head and torso in the VEIL, there are approximately twenty VEIL-volume changes per hour, minimizing viral accumulation and avoiding CO2 rebreathing.
Devised by Prof. Timothy Morris, M.D., Gopesh Tilvawala, and Prof. James Friend, with other engineering and medical staff from UC San Diego School of Medicine, the Jacobs School of Engineering and the Qualcomm Institute, the healthcare worker can treat patients without requriing intubation.
Below please find a video of the device as explained by Prof. Tim Morris, followed by a video of the device’s design and details by Gopesh.
Gopesh explains the design and phenomena behind the VEIL...
Gopesh Tilvawala • Ernesto Criado-Hildago • Alex Grant • William Connacher • Jessica Wen • Teresa Wen • Brian Gee • Edward Aminov • Jeffrey Sandubrae
Prof. Timothy Morris, MD, & Prof. James Friend, PhD team leaders
Work(s) (the “Work”) by:
COVID-19 Acute Ventilation Rapid Response Taskforce (AVERT) Medically Advanced Devices Laboratory
Department of Mechanical and Aerospace Engineering
Jacobs School of Engineering and the School of Medicine University of California, San Diego
9500 Gilman Drive MC411
La Jolla, CA 92093-0411
This Work is Copyright © 2020 The Regents of the University of California. All Rights Reserved. Permission to copy, modify, and distribute this Work and its documentation for educational, research and non-profit non-clinical purposes, without fee, and without a written agreement is hereby granted, provided that the above copyright notice, this paragraph and the following three paragraphs appear in all copies. Permission to make commercial and/or clinical use of this Work may be obtained by contacting:
Office of Innovation and Commercialization
9500 Gilman Drive, Mail Code 0910
University of California
La Jolla, CA 92093-0910
This Work and documentation are copyrighted by The Regents of the University of California. The Work and documentation are supplied “as is”, without any accompanying services from The Regents. The Regents does not warrant that the operation of the Work will be uninterrupted or error-free. The end-user understands that the Work was developed for research purposes and is advised not to rely exclusively on the Work for any reason.
IN NO EVENT SHALL THE UNIVERSITY OF CALIFORNIA BE LIABLE TO ANY PARTY FOR DIRECT, INDIRECT, SPECIAL, INCIDENTAL, OR CONSEQUENTIAL DAMAGES, INCLUDING LOST PROFITS, ARISING OUT OF THE USE OF THIS WORK AND ITS DOCUMENTATION, EVEN IF THE UNIVERSITY OF CALIFORNIA HAS BEEN ADVISED OF THE POSSIBILITY OF SUCH DAMAGE. THE UNIVERSITY OF CALIFORNIA SPECIFICALLY DISCLAIMS ANY WARRANTIES, INCLUDING, BUT NOT LIMITED TO, THE IMPLIED WARRANTIES OF MERCHANTABILITY AND FITNESS FOR A PARTICULAR PURPOSE. THE WORK PROVIDED HEREUNDER IS ON AN “AS IS” BASIS, AND THE UNIVERSITY OF CALIFORNIA HAS NO OBLIGATIONS TO PROVIDE MAINTENANCE, SUPPORT, UPDATES, ENHANCEMENTS, OR MODIFICATIONS.